Precision test for the Sensirion SHT35 humidity sensor
How to tackle the 7 principles of HACCP in restaurants and food companies with simple technology
General Use & Installation
Water level sensor to measure the water level in a container or well
A Weather Station with API
Effortless groundwater level monitoring
How is Reporter powered?
How Crodeon helps research companies save time with remote monitoring and data logging
Crodeon measures air quality and insulating performance for Total Value Wall
How Sint-Rembert optimises ventilation and CO2 in schools
Weather Station Maintenance
How to install a weather station
10 things to consider when buying a professional weather station
Why temperature control is incomplete without remote monitoring
A weather station with solar panel
A weather station with multiple sensors
How KuNa Fish makes fish processing and packaging a piece of cake
3 Reasons why you need a power failure detector at your construction site
Why you need a freezer power outage alarm in an industrial fridge or cold room
What is the difference between ultrasonic and electromagnetic flow meters?
Planning your onion storage facility: 3 things to consider
How Bert gets the best deal for his onions by using a smart storage system
When do you need a titanium level sensor?
4 Things to look for in a temperature sensor with API
Choosing the right tool for measuring relative humidity in concrete structures
3 reasons why storing records needs remote monitoring
3 reasons why you should monitor your roof garden
The importance of lux in offices: 500 lux
Protecting fruit tree blossoms from freezing with the aid of frost detection
Comparing the Watermark sensor to the WaterScout SM100
Pseudomonas syringae and frost damage in fruit trees
Understanding the difference between Crodeon and other remote monitoring systems
Preventing frost damage in a cactus greenhouse
Monitoring heat treatment in cargo containers
4 fantastic ways to monitor your hydro garden or vertical farm
Making blueberry farming more sustainable: Reporter and the water balance model
Measuring lux in chicken pens, pigsties and other stables
PAR & plants, is it the right light sensor for you?
Temperature registration for HACCP
Poultry monitoring and alarm
An easy and reliable weather station
Easy online dewatering
Livestock heat and suffocation alarm
Cold room monitoring
Smart watering system using remote irrigation monitoring
Storm & scaffolding: use a weather station to your advantage
Using a weather station on your construction site
Irrigation control with remote monitoring
Professionally farming mealworms: Measuring CO2, temperature and humidity
Remote Monitoring IoT
We recently tested and documented the precision of the SHT35 T/RH sensor from Sensirion. In this article we explain the results of our tests.
The SHT35 is the high-end version of the SHT3x series that Sensirion calls the new industry standard in humidity sensing. In our case we’re talking about the SHT35-DIS-F version of the SHT35, which includes an on-chip PTFE foil membrane that allows for sensor use under harsh environmental conditions.
What does the datasheet say?
The datasheet of the SHT35 lists the accuracy of all versions of the sensor. Please note that the accuracy is not the same thing as the precision of a sensor. More about this later.
The SHT35-DIS-F offers the highest accuracy of all: 1,5% humidity @0-80% RH) and 0,1°C temperature (@20-60 °C), which is quite impressive. Let’s see if this is the case if we compare multiple of these sensors in real life.
Results of our tests
We tested the precision of 6 of these sensors over a period of 12 hours. The goal was to compare the values with the average of all measurements and see if the deviation was within the accuracy limits.
A few words about the form factor of the sensors. We used these sensors from Crodeon, which are enclosed in a radiation shield suited for outdoor use. All sensors were connected to our Reporter, a wireless gateway with gsm connection to the cloud.
After 12 hours we downloaded the data from the Crodeon Dashboard to Excel. This is where the interesting part started. The following graph shows the temperature readings of our 6 sensors on an interval of 5 minutes. The Reporter can handle more intense intervals up to 30 seconds but for this test we thought 5 minutes was sufficient.
As you can see in the data, almost all of the readings fall within the accuracy limits of -0,1°C and + 0,1°C as claimed in the datasheet.
Now it’s time to investigate the humidity readings. The SHT35 is one of the most accurate humidity sensors on the market. It’s really hard to find this kind of sensors with an accuracy of +/-1,5%RH or better. The following graph did not disappoint us.
Precision versus accuracy
The attentive readers will surely point out that we haven’t really investigated the accuracy of our sensor. That’s right. As explained by Wikipedia, in a set of measurements, accuracy is closeness of the measurements to a specific value, while precision is the closeness of the measurements to each other.
A next quality control would be to test these sensors in a calibrated environment to see how they compare to real world values, which must be measured by a primary of secondary calibrated standard. This is something we will blog about in the near future.
As you can see, we were very satisfied with the performance of the SHT35 digital TRH sensor. The precision is outstanding and we have high hopes for the accuracy as well. We will continue to use this sensor in our agriculture, industry, hvac and research projects.
As Co-Founder & CEO at Crodeon, Jonathan has a broad knowledge about how sensors are changing our daily lives. Over the past 7 years he has been talking to customers in many sectors, from agriculture to industry and everything in between. Jonathan knows the ins and outs of Internet of Things and what keeps our customers awake at night.